1. Fast waveform capture rate and super display ability
The application of digital fluorescence display technology enables DPO to simultaneously display multiple images of signals in different shades or colors. DPO clock can record 200,000 waveforms per second, and its signal data is 1000 times more than ordinary DSO, and each time can capture 500,000 waveforms. This fast waveform acquisition rate combined with excellent display ability enables DPO to analyze any detail of the signal.
2. Continuous high speed sampling capability
Usually DSO for processing display data in the display between two waveforms have 8 ms dead time, even if the sampling technologies adopted instavuTM DSO this time also can reduce to 1.7 mu s, ART cannot capture the waveform oscilloscope in flyback time information and DPO can always at the highest sampling rate of continuous sampling of hundreds of thousands of waveforms, overcome the other between stagnation into the problems of the oscilloscope. The sampling rate of DPO generally has several 109 times per second, which allows oscilloscope to have larger bandwidth.