Bandwidth, class parameter
Generally speaking, the sampling rate is 5 times the bandwidth, such as 200M oscilloscope, with 1G sampling rate is ok.
The pursuit of higher sampling rate is nothing more than to catch small burrs, but these high frequency burrs have been filtered out in the bandwidth layer, and higher sampling rate cannot bring good benefits.
Normal trigger, it will sample the waveform only with trigger event satisfies the condition.
Automatic trigger, after wait for a moment, if there is no eligible trigger events occur, the oscilloscope will force the trigger function and start sampling. As random trigger position, the waveform will show unsteady. If above sequence occurs, please adopt normal mode.
You could catch the signal with template trigger.
When the template trigger is turned on, the template actually comes as a layer. It will constantly detect whether there is waveform that will touch the area of the template. When there is waveform that touches the template, a signal will be detected and it will be filtered and displayed.
No. Bottom of oscilloscope is connected with the earth, it could not be connected directly with the zero line. To isolation, transformer could directly measure 220V city power, however, this is not recommended, and the safest and correct method it to use difference probes.
Two important points Different attenuate bandwidth, signal x1 gear amplitude does not decay. X10 gear signal decay to 1/10 of original signal.
Different bandwidth, bandwidth of X1 gear is almost 10M, and 250M for X10 gear.
The standard does not have a specific metering specification for the probe, but for passive probes, at least the probe compensation must be adjusted when the probe is replaced and the probe switches channels.
The main determinants of price are, of course, bandwidth and functionality.
There are many types of oscilloscope probe, different performance, such as high voltage, difference, active high-speed probe and so on, the price also from a few hundred yuan to nearly ten thousand dollars.
The probe is part of the oscilloscope contact circuit. A good probe can provide the fidelity required for testing.To do this, even passive probes must have a very large number of passive device compensation circuits (RC networks) inside.
Can rely on the oscilloscope's own power analysis software to do, there is a function - B-H curve analysis, it can reflect the working state of the magnetic core, but also can measure the dynamic inductance value, and get the core loss.
Modern digital oscilloscope mostly has FFT function, but need to pay attention to FFT can analyze the number of points, which directly determines the accuracy of FFT analysis results, avoid spectrum leakage.ZDS2000 series oscilloscope is equipped with 4M point FFT analysis
Oscilloscopes typically have a 20MHz bandwidth limit, and this is a hardware filter.Some oscilloscopes also support software filters with adjustable cutoff frequencies.The ZDS2024Plus is adjustable from 100Hz to 100MHz.