GW Instek MSO-2102EA 100 MHz, 2-Ch. Digital Storage Oscilloscope
100 MHz bandwidth
Equipped with a 16-Channel Logic Analyzer and a dual channel 25 MHz arbitrary waveform generator
Real time sample rate for each channel is 1GSa/s
Free Frequency Response Analyzer Software (Download from manufacture website only)
Maximum 10M memory depth and VPO waveform display technology
Waveform update rate up to 120,000 wfms/s
8" WVGA TFT LCD screen display
Maximum 1M FFT provides higher frequency domain resolution measurements
High Pass, Low Pass and Band Pass Filter Functions
29,000 segmented memory sections and waveform search function
I2C/SPI/UART/CAN/LIN serial bus trigger and decoding functions
Data log function is able to track signal changes up to 100 hours
Network storage function
Are the different oscilloscope probes interchangeable?
It's interchangeable as long as it's the same interface
Note, however, that the matching capacitance of the probe matches that of the oscilloscope
So maybe it needs to be adjusted
The other is impedance matching of the probe.The impedance matching part of the probe should be adjusted prior to use.Usually there is an adjustable capacitance at one end of the probe near the oscilloscope, and some probes are near
The probe also has an adjustable capacitance at one end.They are used to adjust the impedance matching of the oscilloscope probe.If the impedance does not match, the measured waveform will be distorted. The method of adjusting the impedance matching of oscilloscope probe is as follows: firstly, the input selection of oscilloscope is played on GND, and then the Y-axis displacement knob is adjusted to make the scan line appear in the middle of the oscilloscope. Check if the scan line is horizontal (i.e., if it coincides with the horizontal center line of the oscilloscope), if not, adjust the horizontal balance knob (usually the analog oscilloscope has this adjustment terminal, in the hole, need to use a screwdriver into the adjustment.Digital oscilloscope need not be adjusted).
Then, the input selection of the oscilloscope is connected to the dc coupling, and the oscilloscope probe is connected to the test signal output end of the oscilloscope (generally, the oscilloscope has this output terminal, usually the square wave signal of 1KHz), and then the scanning time knob is adjusted to make the waveform display for about 2 cycles. Adjust the Y-axis gain knob so that the peak - peak of the waveform is about 1/2 screen width.Then look at the top and bottom sides of the square wave to see if they are horizontal.If the phenomenon of overshoot and tilt occurs, it means that the matching capacitance on the probe needs to be adjusted.Adjust it with a small screwdriver until the waveform on both sides is level and there is no overshot.Of course, due to the quality of the oscilloscope probe, may not be adjusted completely without distortion effect, then can only be adjusted to the best effect.
Principle of handheld oscilloscope
A hand-held oscilloscope USES a narrow beam of high-speed electrons to produce tiny points of light on a screen coated with fluorescent material. Under the action of the measured signal, the electron beam ACTS like the tip of a pen and can draw the curve of the instantaneous value of the measured signal on the screen.The use of oscilloscope can observe various different signal amplitude changes with time waveform curve, but also can use it to test a variety of different electricity, such as voltage, current, frequency, phase difference, adjustment amplitude and so on.
How to choose handheld oscilloscope?
Select handheld oscilloscope mainly pay attention to several indicators, bandwidth, storage depth, sampling rate, waveform update rate, isolation channel, display, language, easy operation, and some other kinetic energy (waveform recording, 232/422/485/CAN/I2C/SPI bus decoding) bandwidth: bandwidth is 100M, 200M commonly now, since buy at least buy 100M, each manufacturer is about the same; Storage depth: this various manufacturers vary from 10K to dozens of M have, in general, a few hundred K is enough, too small waveform expansion can not see, square wave are changed into triangular wave. Sampling rate: as long as this index reaches more than 5 times of the bandwidth, all manufacturers have basically reached; Waveform update rate: this index is also a very important index, waveform update rate is too low will lead to some abnormal signals can not be seen; Isolation channel: in the handheld oscilloscope, this is very important, can reduce the risk of accidental short circuit, electric shock, etc., can also reduce the parasitic capacitance during testing, etc., but fortunately, most manufacturers are isolated.