8560EC spectrum analyzer 30 HZ-2.9 GHZ
Fast digital resolution bandwidths of 1 Hz to 3 MHz
Portable, MIL-PRF-28800, Class 3 rugged
Easily transfer screen image or trace data to PC with BenchLink software
Packaging & Shipping
Pack the instrument with PE bubble bag
Foam-in-place to protect the instrument
Fasten the carton with Belt
Pack the carton with PE film to waterproof
All values are in USD and do not include the customs duties & taxes and other surcharges.
The instrument will ship out within 5 days upon payment received.
We usually choose FedEx/DHL International Economy. Other shipment way is also acceptable
30days warranty, No ROR
During the warranty date, the buyer can return the item if have quality problem.
If you want to return the instruments, the return item has to be the original packing. Without my permission, the instrument cannot be dismantled.
You are responsible for returning the item and paying for return shipping. The payment for the freight cost will not return. Only payment for the instrument will be refund.
The refund will complete within 7 working days after we receive the return items
Analog technique that determines the performance height of an oscilloscope
Although the oscilloscope is not instrument on the top of the hardware requirements, given the oscilloscope is the most familiar with many engineers is one of the biggest single instrument of test equipment market segment, so we interviewed in the world to do the oscilloscope bandwidth GHz level several representative, from the analysis on the oscilloscope hardware, with everyone together to understand the core of the oscilloscope design unit.
Bandwidth, sampling rate and storage depth, is the most intuitive decision an oscilloscope market value of the three features, including bandwidth can reflect the performance of the oscilloscope is the most obvious indicators, sampling rate values and bandwidth have been linked, while the two numerical oscilloscope is directly related to the final price, its value is basic it is decided by simulation unit of hardware performance.
Causes of damage to oscilloscope current probe
The reasons for the damage of the current probe, the methods for preventing the damage and the instructions for using the current probe can be summarized as follows:
Circuit board damage caused by plugging current probe after current amplifier is turned on.
Protection against damage: - do not plug or unplug the current probe while it is still active.
Magnetic ring is fragile material, drop or use too hard when it is easy to damage. Damaged/damaged magnetic rings will result in inaccurate or no longer detectable current.
How to prevent damage and use: - avoid dropping or using too hard.
Magnetic ring coil is thin, over current will cause coil burn out.
Prevent damage method: - avoid the load flow when using.
The current clamp is not aligned, and cracks will make the test inaccurate or unable to measure the current. Be careful how you push the clip.
Method to prevent damage: - when using the current clamp to align. Be careful and be careful when pushing the clip.
The cable can be easily damaged if it is pulled too hard or twisted.
Prevention of damage: - do not use the cable too hard to pull, twist, etc.