8560EC spectrum analyzer 30 HZ-2.9 GHZ
Fast digital resolution bandwidths of 1 Hz to 3 MHz
Portable, MIL-PRF-28800, Class 3 rugged
Easily transfer screen image or trace data to PC with BenchLink software
Packaging & Shipping
Pack the instrument with PE bubble bag
Foam-in-place to protect the instrument
Fasten the carton with Belt
Pack the carton with PE film to waterproof
All values are in USD and do not include the customs duties & taxes and other surcharges.
The instrument will ship out within 5 days upon payment received.
We usually choose FedEx/DHL International Economy. Other shipment way is also acceptable
30days warranty, No ROR
During the warranty date, the buyer can return the item if have quality problem.
If you want to return the instruments, the return item has to be the original packing. Without my permission, the instrument cannot be dismantled.
You are responsible for returning the item and paying for return shipping. The payment for the freight cost will not return. Only payment for the instrument will be refund.
The refund will complete within 7 working days after we receive the return items
Causes of damage to oscilloscope current probe
The reasons for the damage of the current probe, the methods for preventing the damage and the instructions for using the current probe can be summarized as follows:
Circuit board damage caused by plugging current probe after current amplifier is turned on.
Protection against damage: - do not plug or unplug the current probe while it is still active.
Magnetic ring is fragile material, drop or use too hard when it is easy to damage. Damaged/damaged magnetic rings will result in inaccurate or no longer detectable current.
How to prevent damage and use: - avoid dropping or using too hard.
Magnetic ring coil is thin, over current will cause coil burn out.
Prevent damage method: - avoid the load flow when using.
The current clamp is not aligned, and cracks will make the test inaccurate or unable to measure the current. Be careful how you push the clip.
Method to prevent damage: - when using the current clamp to align. Be careful and be careful when pushing the clip.
The cable can be easily damaged if it is pulled too hard or twisted.
Prevention of damage: - do not use the cable too hard to pull, twist, etc.
Basic principle and measuring method of oscilloscope
Oscilloscope is one kind of electrical measurement equipment with wide application. It could transfer invisible electrical signal to visible image to facilitate people research change processes of all kinds of electricity phenomenon. Oscilloscope use narrow electron beam which consists high-speed electronics and project onto screen coated with fluorescent substance to produce small spot (this is original working principle of analog oscilloscopes. Under the function of measured signal, electronics like the point of a pen which could on screen describe change curve of measured signal during instantaneous value. An oscilloscope can be used to observe the waveform curves of various signal amplitudes varying with time. It could also be used to measure various electricity, such as voltage, current, frequency, phase, amplitude modulation and etc.
(1) pre-adjusting: counterclockwise rotate the brightness button to the bottom, shift vertical and horizontal positions to the middle, set the attenuation gear at the highest level, place scan “outside X gear”
(2) turn on the power, do following steps waiting for one or two minutes after the indicator light is on
(3) adjust luminance first, then adjust the focus, adjust vertical and horizontal displacement to make the light point in center of suitable area.
(4) adjust scan, scan fine tuning and X gain, observe scan.
(5) adjust outside X gear to suitable scan range, observe the voltage waveform according to sine and cosines law on vertical direction inside the device.
In general, the frequency of the calibration signal is 1kHz and the amplitude is 0.5v, which is used to calibrate the internal scanning oscillator frequency of the oscilloscope. If it is abnormal, the corresponding potentiometer of the oscilloscope (internal) should be adjusted until it is consistent.